Thank You Sponsors!

AVENSYS.COM

CANCOPPAS.COM

CGIS.CA

ELECTROZAD.COM

EVERESTAUTOMATION.COM

HCS1.COM

PEP-PETRO.COM

SWAGELOK.COM

VANKO.NET

WAJAX.COM

WESTECH-IND.COM

WIKA.CA

## AutoQuiz: Level Reading with a Differential Pressure Cell

The post AutoQuiz: Level Reading with a Differential Pressure Cell first appeared on the ISA Interchange blog site.

AutoQuiz is edited by Joel Don, ISA’s community manager.

Today’s automation industry quiz question comes from the ISA Certified Control Systems Technician (CCST) program. Certified Control System Technicians calibrate, document, troubleshoot, and repair/replace instrumentation for systems that measure and control level, temperature, pressure, flow, and other process variables. Click this link for more information about the CCST program.

A differential pressure (d/p) cell is mounted 3 feet below the zero level in a tank. For a true level reading, the transmitter output must be:

a) calibrated for elevated zero
b) calibrated for span
c) calibrated for true zero
d) calibrated for suppressed zero
e) none of the above

<span class="collapseomatic " id="id3510" tabindex="0" title="Click Here to Reveal the Answer” >Click Here to Reveal the Answer

If the d/p cell is not located at an elevation that corresponds to 0% level in the tank, we must calibrate it to account for the difference in elevation.

This calibration adjustment is “zero elevation” when the cell is located above the lower tap; it is “zero suppression” when the cell is located below the lower tap.

To calculate the amount of elevation or suppression, we must measure the distance between the zero reference level and the high-pressure port of the d/p cell. Taking into account that 27.68″ H2O = one (1) psig, calibrate the transmitter for the distance measured in the direction necessary to result in an accurate output.

When the d/p cell is at the same elevation as the zero level in the tank, calibrate the transmitter for “true zero.”

The correct answer is D, calibrated for suppressed zero.

Reference: Mike Cable, Calibration: A Technician’s Guide, ISA Press.

Source: ISA News